Overall Dimension: - 2035 mm L X 530 mm W X 870 mm H (Approx.)
Construction:- Frame should be made up of 50 mm X 25 mm X 18G CRCA rectangular tube, upper horizontal support tube should be made up of 25 mm X 25 mm X 18G tube. Four no’s 150 mm diameter X 40 mm W heavy duty Swiveling casters bolted to an outer pipe of 31.75 mm diameter.
Detachable stretcher top should be 304 Grade Stainless Steel sheet having 1 mm thickness and concave surface of depth 25 mm, stainless steel handles of 25.4 mm diameter and four legs of 31.75 mm diameter pipe with rubber shoes.
Finish: All components should be pretreated in separate eight tank process
for better finish, good adhesion and corrosion protection. Process includes Hot Degreasing, Derusting, Activation, Phosphating & No’s of Water rinses as per IS 3618 - 1966 class ‘C’ type and then pretreated materials is coated with epoxy powder with film thickness of 60 microns (approx.) and then oven baked at 180 degree centigrade.